- Piezometers (8)
- Inclinometers and Tiltmeters (25)
- RDS Railway Deformation System (6)
- Pendulums (1)
- Settlement Gauges (9)
- Pressure cells (5)
- Load Cells (3)
- Extensometers (1)
- Crackmeters and Jointmeters (1)
- Strain Gauges and Thermometers (3)
- Readout units and dataloggers (48)
- Cables, Accessories and Software (2)
The Barometer model 0MEPR106000 have a 4-20mA current loop output.
The measuring range is from 800 mbar (equivalent to 4 mA) to 1200 mbar (equivalent to 20 mA).
Manual readings are taken connecting the wires to a readout according to the following instructions:
Signal: 4-20mA current loop
Red: + Loop
Black: - Loop
The following formula allows to convert the electrical measurements (mA) into engineering values (mbar):
Pmbar = Pressure in millibar
RmA = Reading in mA
S = sensitivity as from the Compliance Certificate in mA/mbar (usually 25 mbar/mA)
The cable lengths depend from many factors, first of all installation and cable protections (screen, earth, etc...). Assuming that instruments are installed in a workmanlike, for the maximum cable lengths should be followed the next suggestions:
DIGITAL INSTRUMENTS: see FAQ#073
ANALOGUE INSTRUMENTS: see both documents:
Cable length guide line with internal multiplexers
Cable length guide line with external multiplexers
FAQ#037 - Which is the linear thermal expansion coefficient of the graduated flat cable (tape) of Water Level Indicators (WLI) and BRS settlement probes?
The linear thermal expansion coefficient of the flat graduated cable is 16x10E-6 m/ °C.
FAQ#033 - I have received the calibration Report of a piezometer. There is not the K Thermal Factor noted. In which way may I proceed?
If the value is not indicated in the Calibration Report, you should use this formula: K = -1 digit/°C For each degree of increase of temperature you should reduce 1 digit.
HAE (High air entry) filters are normally used to monitor pore pressure in the clay core dams and saturated low permeability soils. “High air entry” means high-resistance to air entry when the filter is saturated. The surface tension of the water in the very fine porus of the filter prevents entry of air. The HAE filters must be saturated before installation by means of the saturation device.
LAE (Lowair entry) filters are often used in medium and soft soils. They do not need to be saturated by special device because the porosity of the filter permits water entry with simple immersion.
A plasma surge arrestor is wired in parallel with the coil (which is used to pluck the wire). The nominal impulse discharge current 8-20 microsec interval is 10 KA. Impulse spark-over voltage at 1Kv/ microsec: 700 V.
FAQ#017 - Can we buy spare parts for replacement of the water level indicator probe and repair it ourselves?
Yes, we sell a kit which includes instructions, probe, weights, rexin, and any other parts.
FAQ#016 - Are the vibrating wire piezometers equipped with temperature sensors, or is the temperature sensor an option?
Yes, each vibrating wire piezometer, as every Sisgeo vibrating wire instrument, is equipped with a built-in NTC thermistor YSI 44005 type (3000 Ohm at 25°C). Thermistor is not a option for vibrating wire instruments and therefore there is not additional cost for it.