Codes der Referenzprodukte: L2M0, L2E0 Hydraulische Ankerkraftmessdosen werden zur Messung der Kräfte in Erd- und Felsankern verwendet. Die Messdosen bestehen aus zwei ringförmigen Platten aus rostbeständigem Stahl die verschweisst werden und im Zwischenraum unter Vakuum mit Öl gefüllt ein Druckmesskissen bilden. Der Druck wird mit einem, an das Druckmesskissen angeschlossenen, Manometer gemessen und direkt in […]
Under EU financing, a Center for monitoring of impact of infrastructure on environment has been created within Technical University of Bucharest.The project was developed though an International Tender which was awarded to SISGEO. The aim was to provide Technical University with comprehensive list of geotechnical equipment, in order to allow University specialist to deal with […]
Sisgeo, thanks to the collaboration with its Russian partners AGT Systems and GPIKO, furnished and installed all the equipments necessary to monitor the most important road and railway works that are under construction in Sochi area, that is the neural centre of 2014 Winter Olympics.The monitored works are: retaining walls for the protection of roads, […]
The cleaning-up project of Panama City and Bay includes: Construction of Health Service networks Construction and rehabilitation of Collectors and Drive lines Interceptor system Plant for wastewater treatment. These elements will be developed to collect, transport and treat wastewater of Panama City which is currently discharged directly into the bay or rivers and streams in […]
It is not possible to repair/change the manometer without disassemble the load cell. The scope of the glycerine inside of the manometer display is only to show eventual tampering of the manometer. The load cell work properly without glycerine, too.
FAQ#015 – The anchor load value displayed on the load cell manometer or readout does not agree with load calculated by the hydraulic jack. What’s right?
The main factors which influence the anchor measurements are: – axial misalignment of the jack– difference in diameter between the jack and the load cell. These factors cause bending of the bearing plate between the load cell and the jack and affects the response of the cell. To minimise this effect use the genuine distribution […]