Frequently asked questions
It is not possible to repair/change the manometer without disassemble the load cell. The scope of the glycerine inside of the manometer display is only to show eventual tampering of the manometer. The load cell work properly without glycerine, too.
The main factors which influence the anchor measurements are: – axial misalignment of the jack– difference in diameter between the jack and the load cell. These factors cause bending of the bearing plate between the load cell and the jack and affects the response of the cell. To minimise this effect use the genuine distribution […]