It could be a problem connected to the power supply given by the readout or logger. When the power supply is in the current loop (2-wire gauge) it is necessary to consider the effect of voltage drop across the shunt resistor on the voltage applied to the gauge. For example, suppose a 4-20mA gauge requires […]
A plasma surge arrestor is wired in parallel with the coil (which is used to pluck the wire). The nominal impulse discharge current 8-20 microsec interval is 10 KA. Impulse spark-over voltage at 1Kv/ microsec: 700 V.
FAQ#016 – Are the vibrating wire piezometers equipped with temperature sensors, or is the temperature sensor an option?
Yes, each vibrating wire piezometer, as every Sisgeo vibrating wire instrument, is equipped with a built-in NTC thermistor YSI 44005 type (3000 Ohm at 25°C). Thermistor is not a option for vibrating wire instruments and therefore there is not additional cost for it.
Removable pressure transducers Reference products codes: PK45C, P252C The removable pressure transducer is designed to measure pore-water pressure through a special Casagrande filter tip. This unique design is especially suitable for long-term monitoring since it allows the instrument to be withdrawn for calibration checks. The removable pressure transducer can be supplied with either vibrating wire or piezo-resistive technology. APPLICATIONS • Long term […]