Make sure ALL the following steps are verified: Sisgeo successfully uses the mobile phone model “BLACKVIEW BV4900 PRO” for all its tests with WR-Log system
Make sure that ALL the following steps are checked:1. The batteries of the mobile phone/tablet and the MIND readout are charged2. You have installed the latest version of the MIND APP (check the Google Play Store)3. The version of the mobile phone/tablet’s operating system is Android 10 or higher4. The Bluetooth version of the mobile […]
FAQ#119 – Why can’t I connect the BRAIN reel to my mobile phone/tablet? Have I connected correctly my BRAIN measuring system?
Make sure that ALL the following steps are checked:1. The batteries of the mobile phone/tablet and the BRAIN system are charged2. You have installed the latest version of the BRAIN APP (check the Google Play Store or Apple Store)3. The version of the mobile phone/tablet’s operating system is Android 7 or higher, or iOS 11 […]
FAQ#118 – Is it possible to use the reference reading (zero reading) taken with a digital inclinometer probe as a reference reading for another digital inclinometer probe?
No, this is not possible.Sisgeo calibrates each probe individually, but each probe has its own small different properties, that have a certain influence on the readings. The international standard ISO 18674-3, describing the reference reading, states that:“It is good practice to take an additional reference measurement with an independent probe that can be used as […]
Preliminary verifications: • B.R.A.IN reel’s battery is charged• Your mobile or tablet has the following minimum features: Bluetooth Low Energy BLE 4.2 or higher Android OS 7 or higher If the preliminary verification is positive, please proceed as follows:• Un-install the B.R.A.IN app from your device (if […]
Sisgeo digital instruments can operate in two powering modes: TIMED or ALWAYS-ON (for more information see F.A.Q.#094.A string of mixed instruments consisting of TIMED gauges and ALWAYS-ON gauges cannot work.The first thing to do is therefore to check that ALL connected instruments in your array are set to TIMED mode.You can check the powering mode […]
The OMNIAlog logger for FLX-Rail has a fairly high power consumption due to the router.If OMNIAlog is set to send data through the router after each acquisition (so every 5 minutes) it needs to be powered from the network (e.g. 220Vac)If OMNIAlog is set to send data through the router once a day, it can […]
Verify that the mobile phone/tablet used does have the minimum required features:Bluetooth Low Energy BLE 4.2APPLE iOS 14 or higherAndroid OS 9 (suggested Android OS 10 or higher) Then please check that the mobile phone/tablet has Bluetooth on and that the MIND unit has a sufficiently charged battery.
The up-date troubleshootings are available within MIND readout User Manual (see relevant DOWNLOD area or product page). The main troubleshootings are:
It could be a problem connected to the power supply given by the readout or logger. When the power supply is in the current loop (2-wire gauge) it is necessary to consider the effect of voltage drop across the shunt resistor on the voltage applied to the gauge. For example, suppose a 4-20mA gauge requires […]
FAQ#110 – What’s the purpose of the linear and polynomial factors written in the Calibration Reports?
Utilize the Linear Sensitivity Factors (A, B) and Ploynomial Sensitivity Factors (A, B, C, D) of the Calibration Reports permit to obtain readings in engineerig units with a maximum error as for the Calibration Report. In the case of analogue gauges, the factors shall be applied on the electrical readings (i.e. mA, mV, digit, etc…) […]
FAQ#109 – Which are the limitations in terms of instruments that can be connected to WR-LOG Digital Nodes?
The limitations in terms of instruments to be connected on WR-LOG digital nodes are the following: